|Metering Units represent the secondary values for the current and voltage transformers. That is Metering Units are the values present at the voltage and current inputs at the back of the meter.
As an example, if the PTs (voltage transformers) have a ratio of 480:120 and the CTs (current transformer) have a ratio of 300:5; and you were injecting full secondary values (120 Vl-l and 5 A) for all three phases you would see the the following values:
kw A = 120 V * 5 A = 600 W = 0.600 kW
kw B = 120 V * 5 A = 600 W = 0.600 kW
kw C = 120 V * 5 A = 600 W = 0.600 kW
kw total = 0.600 kw + 0.600 kw + 0.600 kw = 1.80 kw
*note: these calculation assume a unity power factor
*note: Metering Units can sometimes be referred to as "uncompensated" or "unscaled" values.
Last Revised: October 3, 2012
All content © 1992-2012 Schneider Electric
Legacy KB System (APS) Data: RESL188299 V3.0, Originally authored by KBAdPM on 10/04/2007, Last Edited by MaTh on 10/10/2012
Related ranges: ION8800, ION8600, ION7550/ION7650, ION7300